Notes from the Field: Mycobacteria chimaera Infections Associated with Heater-Cooler Unit Use During Cardiopulmonary Bypass Surgery — Los Angeles County, 2012–2016
Weekly / January 4, 2019 / 67(5152);1428–1429
M. Claire Jarashow, PhD1,2; Dawn Terashita, MD1; Sharon Balter, MD1; Benjamin Schwartz, MD1 (View author affiliations)View suggested citation
In December 2016, hospital A in Los Angeles County, California, reported two Mycobacterium avium complex infections, later identified as Mycobacterium chimaera, in patients with a recent history (<5 years) of cardiopulmonary bypass surgery. Both surgical procedures used the Sorin Stöckert 3T (Sorin Group, Munich, Germany) heater-cooler unit brand (currently LivaNova PLC, London, United Kingdom) to heat and cool blood. These heater-cooler units have been linked to outbreaks of M. chimaera infections among patients with similar surgical histories in Europe and the United States (1,2). Sorin Stöckert 3T heater-cooler units contaminated during manufacturing before September 2014 were identified as the source of infection through emission of bioaerosols containing M. chimaera during surgery (3); these units have been removed and replaced by hospital A.
M. chimaera is a nontuberculous mycobacterium first described in 2004 (4). M. chimaera infection diagnosis is challenging because clinical manifestations can take months or years to develop and are often nonspecific. Infections have been diagnosed up to 6 years after initial surgical exposure (5). Acid-fast bacillus cultures might not be ordered, or results might be negative given the slow-growing nature of M. chimaera (5,6). In hospitals with confirmed M. chimaera infections, reported incidence rates among heater-cooler unit–exposed patients ranged from one per 100 persons to one per 1,000 persons (2,5), and the case-fatality rate was approximately 50% (6,7). Infections were reported most frequently among patients who had valve replacement or other implants during surgery (8).
CDC released a health alert in October 2016 recommending that hospitals that used Sorin Stöckert 3T heater-cooler units notify patients who were potentially exposed during 2012–2016. Because hospital A used implicated heater-cooler units, an investigation was initiated by the Los Angeles County Department of Public Health in December 2016, to enhance case findings and implement control measures. During the investigation, approximately 4,000 patients were sent letters per CDC guidance, describing the potential exposure and instructing them to seek care if they experienced signs or symptoms consistent with M. chimaera infection, such as fatigue, unexplained fever, night sweats, weight loss, or wound infection. A nurse call center was established to answer patient questions and refer to care when necessary. All relevant clinical staff members were notified, and an alert was inserted into electronic health records of potentially exposed patients. Hospital A was advised to report all M. chimaera cases to the Food and Drug Administration via MedWatch.
By May 2017, 20 confirmed cases of M. chimaera infection had been identified, defined as isolation of culture-positive nontuberculous mycobacterium from an invasive nonpulmonary specimen, with M. chimaera species identification by DNA sequencing of 16S rRNA, in a patient with a history of cardiopulmonary bypass during 2013–2016. Fifteen (75%) cases were identified by clinicians during patient hospitalization, follow-up care, or subsequent surgical procedures at hospital A or affiliated facilities. Five (25%) patients sought care because they received a patient notification letter and subsequently received a diagnosis of M. chimaera infection. All five patients identified through patient notification letters had valve replacements or implants inserted during surgery, and all five remain alive. Thirteen of the 15 patients identified during hospitalization, follow-up care, or subsequent surgery had valve replacements or implants, and eight of these 15 patients were alive at the time this report was produced.
Informing and reminding exposed persons to seek care for M. chimaera–associated nonspecific symptoms can be important for diagnosis, particularly because subsequent care might not occur at the exposure hospital, limiting the likelihood of complete exposures being known. Because of M. chimaera’s long incubation time, hospitals that used implicated heater-cooler units could consider additional proactive steps toward early detection of infection, such as annual patient renotification and implementation of clinician alerts in electronic medical records.
Moon Kim, Los Angeles County Department of Public Health, California; Hector Rivas, Nicole Green, Public Health Laboratory, Los Angeles County Department of Public Health, California; Kiran Perkins, Matt Crist, Division of Healthcare Quality Promotion, National Center for Emerging and Zoonotic Infectious Diseases, CDC; Erin Epson, California Department of Public Health.
Corresponding author: M. Claire Jarashow, CJarashow@ph.lacounty.gov, 213-288-7049.
All authors have completed and submitted the ICMJE form for disclosure of potential conflicts of interest. No potential conflicts of interest were disclosed.
- Sax H, Bloemberg G, Hasse B, et al. Prolonged outbreak of Mycobacterium chimaera infection after open-chest heart surgery. Clin Infect Dis 2015;61:67–75. CrossRefexternal icon PubMedexternal icon
- Lyman MM, Grigg C, Kinsey CB, et al. Invasive nontuberculous mycobacterial infections among cardiothoracic surgical patients exposed to heater-cooler devices. Emerg Infect Dis 2017;23:796–805. CrossRefexternal icon PubMedexternal icon
- Sommerstein R, Rüegg C, Kohler P, Bloemberg G, Kuster SP, Sax H. Transmission of Mycobacterium chimaera from heater-cooler units during cardiac surgery despite an ultraclean air ventilation system. Emerg Infect Dis 2016;22:1008–13. CrossRefexternal icon PubMedexternal icon
- Tortoli E, Rindi L, Garcia MJ, et al. Proposal to elevate the genetic variant MAC-A, included in the Mycobacterium avium complex, to species rank as Mycobacterium chimaera sp. nov. Int J Syst Evol Microbiol 2004;54:1277–85. CrossRefexternal icon PubMedexternal icon
- Schreiber PW, Sax H. Mycobacterium chimaera infections associated with heater-cooler units in cardiac surgery. Curr Opin Infect Dis 2017;30:388–94. CrossRefexternal icon PubMedexternal icon
- Sommerstein R, Schreiber PW, Diekema DJ, et al. Mycobacterium chimaera outbreak associated with heater-cooler devices: piecing the puzzle together. Infect Control Hosp Epidemiol 2017;38:103–8. CrossRefexternal icon PubMedexternal icon
- Walker J, Moore G, Collins S, et al. Microbiological problems and biofilms associated with Mycobacterium chimaera in heater-cooler units used for cardiopulmonary bypass. J Hosp Infect 2017;96:209–20. CrossRefexternal icon PubMedexternal icon
- Sommerstein R, Hasse B, Marschall J, et al. ; Swiss Chimaera Taskforce. Global health estimate of invasive Mycobacterium chimaera infections associated with heater–cooler devices in cardiac surgery. Emerg Infect Dis 2018;24:576–8. CrossRefexternal icon PubMedexternal icon
Suggested citation for this article: Jarashow MC, Terashita D, Balter S, Schwartz B. Notes from the Field: Mycobacteria chimaera Infections Associated with Heater-Cooler Unit Use During Cardiopulmonary Bypass Surgery — Los Angeles County, 2012–2016. MMWR Morb Mortal Wkly Rep 2019;67:1428–1429. DOI: http://dx.doi.org/10.15585/mmwr.mm675152a4external icon.
MMWR and Morbidity and Mortality Weekly Report are service marks of the U.S. Department of Health and Human Services.
Use of trade names and commercial sources is for identification only and does not imply endorsement by the U.S. Department of Health and Human Services.
References to non-CDC sites on the Internet are provided as a service to MMWR readers and do not constitute or imply endorsement of these organizations or their programs by CDC or the U.S. Department of Health and Human Services. CDC is not responsible for the content of pages found at these sites. URL addresses listed in MMWR were current as of the date of publication.
All HTML versions of MMWR articles are generated from final proofs through an automated process. This conversion might result in character translation or format errors in the HTML version. Users are referred to the electronic PDF version (https://www.cdc.gov/mmwr) and/or the original MMWR paper copy for printable versions of official text, figures, and tables.
Questions or messages regarding errors in formatting should be addressed to email@example.com.