Electromagnetic spatial/spectral sensor for geological measurements.
Proceedings of the Sixth International Conference on Ground Penetrating Radar, Sep 30 - Oct 3, 1996, Sendai, Japan. Sendai, Japan: Tohoku University, Department. of Geoscience and Technology, 1996 Sep; :545-547
The U.S. Department of Energy Pittsburgh Research Center, has developed and field tested a spatial, spectral 1 sensor for measuring the thickness of mineral deposits. Knowledge of the dielectric constant of the material is not necessary because the electrical properties of the media are determined automatically along with the thickness or distance measurement. The technology was developed to measure the thickness of coal remaining in the roof and floor of a mine or the thickness of a coal pillar remaining between two mining entries for the purpose of guiding a mining machine. However, field tests have shown the new technique is equally valuable for measuring the thickness of each layer in a multilayer manmade structure such as highway paving or building foundations and the thickness of geological deposits other than coal, such as, salt, granite and sandstone. The noncontacting electromagnetic technique uses spatial modulation created by moving a simple dipole sensor antenna in a direction along each axes to be measured while the complex reflection coefficient is measured at multiple frequencies over a two-to-one bandwidth. The antenna motion imparts spatial modulation to the data and enables frequency domain signal processing to solve the problem of media, target and antenna dispersion. The technique overcomes the problem of multipath reflections from nearby metal structures and avoids the use of radio frequency absorbing material. In its present form the system measures material thickness of zero to over 3 m to within +/-2.5 cm accuracy. The system parameters are variable to suit the application but for most applications default values of 40 cm antenna motion with measurements made at 401 frequencies from 0.6 to 1.4 GHz at each of 32 antenna positions with a power level of 0 dBm or less are more than adequate to provide a satisfactory signal-to noise ratio. In the range of 0 to 10 m the system operates in the continuous wave mode and by simply turning off the transmitter for ranges beyond 10 m the problems of time domain processing are entirely avoided. The technique can be implemented with a vector network analyzer, a servo controlled L-band antenna positioner, an IEEE488 bus controller and a process computer. An electronics package designed and constructed to operate in the harsh mining environment and providing 1, 2 or 3 dimensional image presentation is being evaluated as a replacement for the network analyzer.
Mining-industry; Underground-mining; Coal-mining; Mining-equipment; Geology; Measurement-equipment; Spatial-perception
Other Occupational Concerns
Proceedings of the Sixth International Conference on Ground Penetrating Radar