NIOSHTIC-2 Publications Search
Determining Impurities in Helium.
Emerson-DE; Weems-GW; Hoffman-CA
U S Pat 3 662 588 1972 May; :
For the mass spectrographic determination of impurities in helium, the impurities were collected from the flowing he sample by solidification in a trap and coil system cooled in a liquid he bath; the He sample is maintained above atmospheric pressure and at a flow rate sufficient to avoid its liquefaction. After evacuation, the trapping assembly is warmed to ambient temperature to vaporize the collected impurities for mass spectrographic determination. A dewar vessel containing the liquid he is maintained at a proper level with respect to the trap and coil assembly by an automatically controlled lift. A detection limit of 0.01 Ppm can be achieved.
U.S. Pat. 3,662,588; May 16, 1972; Chem. Abstr. 77:96,620R
Page last reviewed: April 12, 2019
Content source: National Institute for Occupational Safety and Health Education and Information Division