NIOSHTIC-2 Publications Search
A method for calculating m/e's of peaks from time-of-flight mass spectra.
Appl Spectrosc 1972 May-Jun; 26(3):403-404
This study has shown that m/e's can be determined to the nearest unit mass from tof spectra by using a single equation in conjunction with a calibration curve. The calibration curves are unique for individual analog scanners, are stable within 0.3 Amu over a 3-week time span, and are independent of scan rate within 0.3 Amu even when the scan rate is varied by a factor of 4. The authors feel that a 25-cm record for 200 mass units is minimal, and any shorter record would introduce significant measuring errors. Where the analyst has been using an internal standard he can now make a single run on the marker compound and construct a calibration curve to be used over a period of time.
OP; Journal Article
Issue of Publication
Page last reviewed: March 11, 2019
Content source: National Institute for Occupational Safety and Health Education and Information Division