NIOSHTIC-2 Publications Search
Failure Report No. S325.
NIOSH 1977 Aug:8 pages
Test results using the NIOSH analytical method S325 for particulate fluoride (16984488) dust are reported. The method involves collection on a mixed cellulose ester membrane filter, ashing, fusion with a carbonate mixture, and analysis by a fluoride specific electrode. Sample preparation and ashing procedures are described. Data on recovery rates from spiked samples is included. The method was unsatisfactory due to poor recovery rates and an unacceptably high coefficient of variance.
NIOSH-Sampling-Data-Sheet; NIOSH-Backup-Data-Sheet; Fluorides; Particulate-dust; Membrane-filters; Sampling; Inorganic-salts;
NTIS Accession No.
NIOSH, Unpublished, Cincinnati, Ohio, 8 pages, 11 references
Page last reviewed: September 2, 2020
Content source: National Institute for Occupational Safety and Health Education and Information Division