NIOSHTIC-2 Publications Search
NIOSH Analytical Method, Stibine, S243, (Set 2).
NIOSH 1977 Jul; :1-9
The method is described for determining stibine (7803523) in the air, including the principle, range and sensitivity, interferences, precision and accuracy, advantages and disadvantages, apparatus, reagents, calibration and standards, and calculations. The principle is based upon colorimetry of samples collected on mercuric- chloride coated silica gel after extraction with hydrochloric-acid. The range is 0.119-1.008 milligrams per cubic meter with a precision of 0.087 (coefficient of variation). The OSHA standard is 0.1ppm or 0.5 milligram per cubic meter.
NIOSH-Author; NIOSH-Analytical-Method; Standards-Completion-Program; SCP; Analytical-Method; Standards; Sampling; Set-2; Air-quality-measurement; Air-sampling; Procedures; Exposure-limits; Antimony-compounds; Metal-compounds
Division of Physical Sciences and Engineering, NIOSH, Cincinnati, Ohio, 9 pages, 3 references
Page last reviewed: September 2, 2020
Content source: National Institute for Occupational Safety and Health Education and Information Division