Mining Publication: Method Factors for Anemometer Measurement at Pipe Outlets
Gob holes are routinely used by the mining industry to vent methane from gobs and to prevent it from entering the mine ventilation system. In some cases, flows from gob holes are measured by centering an anemometer on the end of a discharge pipe. These measurements are erroneous and tend to be high by as much as 30 pct. The Bureau of Mines determined method factors (correction factors) for anemometer measurements taken in this manner. Method factors for 4-, 6-, and 8-in (10.2-, 15.2-, and 20.3-cm) ID pipe are 0.68, 0.71, and 0.78, respectively. Comparisons between horizontal and vertical flows indicated no significant change in method factors. Consequently, the method factors for horizontal pipe flows can be applied to vertical upward flows.
Report of InvestigationsJanuary - 1987
NIOSHTIC2 Number: 10005324
U.S. Department of the Interior, Bureau of Mines. Report of Investigations 9061, 1987; :1-10